|Year : 2014 | Volume
| Issue : 11 | Page : 1045-1055
Scanning laser polarimetry in glaucoma
Tanuj Dada, Reetika Sharma, Dewang Angmo, Gautam Sinha, Shibal Bhartiya, Sanjay K Mishra, Anita Panda, Ramanjit Sihota
Dr. Rajendra Prasad Centre for Ophthalmic Sciences, All India Instituteof Medical Sciences, New Delhi, India
|Date of Submission||05-Jul-2014|
|Date of Acceptance||03-Nov-2014|
|Date of Web Publication||10-Dec-2014|
Dr. Rajendra Prasad Centre for Ophthalmic Sciences, All India Institute of Medical Sciences, New Delhi
Source of Support: None, Conflict of Interest: None
Glaucoma is an acquired progressive optic neuropathy which is characterized by changes in the optic nerve head and retinal nerve fiber layer (RNFL). White-on-white perimetry is the gold standard for the diagnosis of glaucoma. However, it can detect defects in the visual field only after the loss of as many as 40% of the ganglion cells. Hence, the measurement of RNFL thickness has come up. Optical coherence tomography and scanning laser polarimetry (SLP) are the techniques that utilize the evaluation of RNFL for the evaluation of glaucoma. SLP provides RNFL thickness measurements based upon the birefringence of the retinal ganglion cell axons. We have reviewed the published literature on the use of SLP in glaucoma. This review elucidates the technological principles, recent developments and the role of SLP in the diagnosis and monitoring of glaucomatous optic neuropathy, in the light of scientific evidence so far.
Keywords: Fixed corneal compensation, glaucoma, retinal nerve fiber layer, scanning laser polarimetry
|How to cite this article:|
Dada T, Sharma R, Angmo D, Sinha G, Bhartiya S, Mishra SK, Panda A, Sihota R. Scanning laser polarimetry in glaucoma. Indian J Ophthalmol 2014;62:1045-55
|How to cite this URL:|
Dada T, Sharma R, Angmo D, Sinha G, Bhartiya S, Mishra SK, Panda A, Sihota R. Scanning laser polarimetry in glaucoma. Indian J Ophthalmol [serial online] 2014 [cited 2020 May 28];62:1045-55. Available from: http://www.ijo.in/text.asp?2014/62/11/1045/146707
Glaucoma is an acquired progressive optic neuropathy characterized by changes in the optic nerve head and retinal nerve fiber layer (RNFL). White-on-white perimetry is the gold standard for diagnosis of glaucoma. Various new technologies are coming up for the early diagnosis of glaucoma. We performed a systemic search of the PubMed using the terms-glaucoma, scanning laser polarimetry (SLP), GDx and RNFL to prepare this review, which elucidates the technological principles, recent developments and the role of SLP in the diagnosis and monitoring of glaucomatous optic neuropathy.
| Glaucoma and its Diagnosis: A Background|| |
Imaging of RNFL is of vital importance in glaucoma as structural RNFL changes often precede functional visual field changes. ,, As many as 40% of all ganglion cells can be lost before a well-defined scotoma is detected on the visual field.  RNFL evaluation has been found to be more sensitive for predicting future visual field loss when compared to ONH evaluation and a better predictor of damage. ,,,,, Recent technology boom in ophthalmology has given us two technologies that are capable of providing highly reproducible RNFL assessments. Optical coherence tomography (OCT) based on low-coherence interferometry, generates posterior segment thickness measurements with an axial resolution of eight to ten microns while SLP provides RNFL measurements based upon the birefringence of the retinal ganglion cell axons.
| Technologic Principle|| |
Retinal nerve fiber layer is made of highly ordered parallel axon bundles which contain microtubules, cylindrical intracellular organelles with diameters smaller than the wavelength of light. The paralleled structure of the microtubules is the source of RNFL birefringence that is the splitting of a light wave by a polar material into two components. These components travel at different velocities that create a relative phase shift termed "retardation". This retardation is proportional to the thickness of the RNFL. , Total retardation of a subject's eye is the sum of cornea, lens, and RNFL birefringence. Compensation of anterior segment birefringence is thus necessary to isolate RNFL birefringence.
Scanning laser polarimetry is basically a confocal scanning laser ophthalmoscope with an integrated ellipsometer to measure retardation. It determines the RNFL thickness, point by point in the peripapillary region, by measuring the total retardation in the light reflected from the retina. Polarized light passes through the eye and is reflected off the retina. ,,, Because the RNFL is birefringent, the two components of the polarized light are phase shifted relative to each other and this is captured by a detector, and converted into thickness (in microns). 
| Generations of Scanning Laser Polarimetry and Calculation of Birefringence|| |
Fixed corneal compensation
The first-generation device (nerve fiber analyzer [NFA] I) became commercially available in 1992 and was equipped with a single detector, which was later replaced by a double detector (NFA II). These earlier versions of SLP (e.g. the GDx NFA, GDx Access) were compensated for anterior segment birefringence based on fixed values for the axis and magnitude of the anterior segment birefringence. However, later it was found that the axis and magnitude are variable for each individual and using a fixed corneal compensation may not adequately account for the aforementioned variables. 
Variable corneal compensation
In 2002, variable corneal compensation (VCC) was introduced, allowing eye-specific compensation of anterior segment birefringence, commercially available in the 5 th iteration of the SLP instrument named GDx-VCC (Carl Zeiss Meditec, Inc., Dublin, California, USA).
GDx-VCC measures and individually compensates for anterior segment birefringence, using a 780-790 nm laser diode. ,, The axis of the anterior segment birefringence is determined by the orientation of the 'bow-tie' birefringent pattern [Figure 1] in the macula and the magnitude of the anterior segment birefringence is calculated by analyzing the circular profile of the birefringence in the macula according to standard equations.  In cases of macular pathology, an alternative method is available that accurately compensates for the anterior segment birefringence. 
|Figure 1: "Bow-tie pattern" (arrow) seen in the macula in an uncompensated scan|
Click here to view
GDx-Enhanced corneal compensation
The enhanced corneal compensation (ECC) algorithm is implemented in the GDx-VCC by a software modification. A known birefringence bias is introduced into the measurement beam path to shift the measurement of total retardation into a more sensitive region of the curve of detection of polarization of the instrument. The bias retarder is formed by the combination of the variable corneal compensator and cornea. However, instead of completely canceling corneal birefringence, the retarder is adjusted and hence that the combination has retardance close to 55 nm and slow axis of polarization close to vertical. After image acquisition, the birefringent bias is removed mathematically, point-by-point, to yield the RNFL retardation values that are converted to thickness (in micrometers) using a fixed conversion factor.
The computerized export of the temporal-superior- nasal-inferior-temporal (TSNIT) plots on the GDx-ECC printout includes the mean RNFL thickness from 64 polar sectors (5.625°/arc). The mean for each sector is computed along the 3.2 mm diameter measurement circle surrounding the optic nerve head. The mean RNFL thickness for the superior (0-180°) and inferior (181-360°) retinal region is computed separately. ECC scans show a stronger structure-function relationship with perimetry compared with VCC. ,
Modified diameter scan
GDx-VCC uses a fixed scan circle of 3.2 mm diameter centered on the optic disc ( conventional diameter scan ) . This may be affected by peripapillary atrophy (PPA). So it has measurement options with two additional scan diameters - medium and large. Dada et al.  found that medium and large diameter scans can also be used to discriminate between normal and glaucomatous eyes. Nerve fiber indicator (NFI) has been seen to be the best discriminating parameter across all diameters.
Retinal nerve fiber layer measurements
The GDx-VCC measurements are taken by scanning the beam of a near-infrared laser (780 nm) in a raster pattern  which captures an image with a field 40° horizontally by 20° vertically, and including both the peripapillary and the macular region.  It generates two images: A reflectance image and a retardation image [Figure 2]. Each image is made up of 256 (horizontal) ×128 (vertical) pixels, or 32,768 total pixels. For an emmetropic eye, 1 pixel is 0.0465 mm in size, and the total scan field is 11.9 mm (horizontal) × 5.9 mm (vertical). 
|Figure 2: Images generated by the GDx-variable corneal compensation: Left image is the refl ectance image displayed as a colored intensity map. The right image is the retardation map converted to retinal nerve fiber layer thickness, color coded based on the color spectrum|
Click here to view
| Clinical Interpretation of the GDx Printout|| |
For each GDx-VCC scan, an age-matched comparison is made to the normative database, and any significant deviations from normal limits are flagged as abnormal with a 'P' value. Quantitative RNFL evaluation is provided through four key elements of the printout [Figure 3]:
|Figure 3: Print out showing various parameters for quantitative retinal nerve fi ber layer evaluation|
Click here to view
|Figure 4: (a and b) Calculation circle, characteristic double hump pattern|
Click here to view
Nerve fiber indicator
The NFI is a global measure based on the entire RNFL thickness map. NFI ranges from 1 to 100, with lower values (<25) indicating a normal RNFL. ,,,
Scanning laser polarimetry technology can help in early diagnosis of glaucoma by picking up RNFL changes that may precede visual field damage by 5-6 years [Table 1].
Age has been found to significantly affect all RNFL measurements with the ECC protocol of SLP, whereas typical scan score (TSS) and residual anterior segment retardance affect the overall average and the superior average RNFL measurements, respectively. 
Medeiros et al.  found GDx-VCC to be superior versus Heidelberg retina tomograph 3 (HRT3) for detecting early damage in glaucoma suspects. They studied 82 glaucoma suspects. The area under the receiver operator characteristic (AUROC) curve for the best parameter from GDx-VCC (NFI) was significantly larger than that of the best parameter from the HRT (rim volume) (0.83 vs. 0.70).
Mohammadi et al.  studied one eye from each of 160 glaucoma suspects with normal standard automated perimetry (SAP) visual fields. They found thinner baseline SLP RNFL measurements to be independent predictors of future visual field damage.
Badalΰ et al.  in their study compared OCT, GDx-VCC, HRT and stereophotographs in 46 healthy subjects and 46 glaucoma patients with early defect. The largest AUROC for each technique was 0.96 for OCT average thickness, 0.92 for GDx-VCC NFI, 0.91 for HRT3 FSM discriminant function and 0.97 for stereophotographs.
Shaikh et al.  studied 39 "glaucoma suspects." Twenty-two of 39 (56.4%) were considered to be at risk of developing progressive glaucoma. In 19 of these patients, abnormal GDx-VCC results were found, particularly inter-eye asymmetry in the RNFL thickness. However, in two of 39 (5.1%) patients the GDx-VCC was normal, despite the presence of a neuroretinal rim defect in the optic disc with corresponding visual field loss, and in one patient with POAG. Hence, they concluded that GDx-VCC should not be used in isolation.
Li et al. found NFI to be the best performing parameter using a cutoff of 35 with a sensitivity of 75%, specificity of 95%, a positive predictive value of 25 and a negative predictive value of 99 for diagnosing glaucoma. 
In a study to assess the relationship between high-definition OCT (HD-OCT) and SLP with VCC and ECC in measuring RNFL in healthy eyes and those with early-to-moderate glaucomatous VF loss, HD-OCT parameters of RNFL thickness were found to be significantly higher than VCC and ECC parameters, and therefore those thickness values are not interchangeable in normal eyes and in early glaucoma patients. 
According to Zheng et al.,  NFI and inferior average are the most effective indicators for the early diagnosis of glaucoma. AUROCs were 0.81 for NFI and inferior average with better differentiation capability.
Da Pozzo et al.  evaluated the diagnostic accuracy of GDx-VCC comparing 62 healthy and 48 glaucomatous age-matched patients with early glaucomatous field defect. The three parameters with largest AUROCs were NFI (0.870), superior average (0.817) and normalized superior area (0.816).
Parikh et al.  evaluated 74 eyes with early glaucoma and 104 eyes of normal subjects. TSNIT SD had the best combination of sensitivity and specificity-61.3 and 95.2, respectively-followed by NFI > 50 (sensitivity, 52.7%; specificity, 99%).
Fortune et al. in their study found that the onset of progressive loss of RNFL retardance occurs earlier than the onset of RNFL thinning. Endpoints of progressive loss from baseline also occurred more frequently and earlier for RNFL retardance as compared with a thickness. 
Reus and Lemij  compared 77 healthy eyes and 162 glaucomatous patients. AUROC for main GDx-VCC parameters ranged from 0.90 to 0.98.
| Ocular Hypertension|| |
Henderson et al.  studied 44 ocular hypertension (OHT) patients and 48 healthy subjects, all of similar age. Higher NFI scores were correlated significantly with thinner CCT measurements in OHT patients. In multivariate analysis, only age and CCT measurement were associated significantly with RNFL measurements in OHT eyes.
Normal tension glaucoma
Choi et al.  evaluated RNFL in the retinal segments without visual field loss in eyes with 56 Asian normal tension glaucoma (NTG) patients who had localized visual field defects confined to 1 hemifield and 43 normal controls. They found that perimetrically normal hemifields of NTG eyes had significantly lower RNFL thickness parameters than the corresponding regions of healthy eyes.
Normal tension glaucoma versus high tension glaucoma
Jung et al.  found GDx-VCC to be more sensitive in detecting RNFL damage in high tension glaucoma (HTG) patients compared to NTG patients. Compared with eyes with NTG, eyes with HTG showed reduced RNFL thickness.
| Juvenile Open-Angle Glaucoma|| |
Zareii  compared the ability of GDx-VCC and OCT to discriminate eyes with Juvenile Open-Angle Glaucoma from normal eyes. They observed statistically significant correlations between the two. The greatest AUROC parameter on OCT (inferior average: 0.92) had a lower area than that in GDx-VCC (NFI: 0.99).
Dimopoulos et al.  compared RNFL thickness of normotensive eyes with exfoliation syndrome (XFS) and healthy eyes using GDx-VCC. Polarimetry-determined RNFLT was lower in XFS eyes with normal IOP. Therefore, close monitoring of RNFLT may facilitate early identification of those XFS eyes that convert to exfoliative glaucoma.
| Detecting Progression|| |
The new software used in GDx that helps in investigating progression is GPA TM (progression analysis for GDx) [Figure 6]. The GDx-GPA algorithms are designed to have 95% specificity for "Likely progression."
GDx-GPA uses two different algorithms:
· Change from baseline (CFB): Based on changes from two baseline exams compared to measurement variability
· Statistical image mapping: Based on trend analysis. All visits contribute to change detection here in contrast to CFB where data from the first two and last two visits are used.
| Atypical Retardation Patterns|| |
Atypical retardation patterns (ARPs) have been characterized as irregular patches of elevated retardation values that are against the expected retardation distribution based on the RNFL anatomy.  ARPs may provide fallacious RNFL measurements  and occur in approximately 10-25% of healthy eyes and 15-51% of glaucomatous eyes, ,, in elderly subjects,  in lightly pigmented fundi and in high myopes. SLP images are generally excluded from the analysis when the software flags them as incompatible with its normative database. The software flags an image as such when the TSS is below 25. 
Typical scan score
TSS is a proprietary measure provided by the GDx-VCC software that indicates whether the observed retardation pattern is typical of the human healthy or glaucomatous RNFL (range: 0-100). In scans that display an ARP, the TSS is lower. It is highly predictive of the presence of ARPs.  In clinical studies, a cut-off value for TSS of 80 is often used to discriminate between normal and atypical retardation patterns. ,, The ability of SLP to discriminate between healthy and glaucomatous eyes has been shown to decrease when ARPs are present. ,, Diagnostic accuracy has been found to be comparable for SD-OCT and GDx-VCC if typical scans (TSS = 100) are investigated.  Decreasing TSS is associated with a decrease in diagnostic accuracy for discriminating healthy and glaucomatous eyes by SLP. NFI has been found to be less influenced than the global or sector RNFL thickness. GDx ECC helps in the neutralization of ARPs.
Wang et al. evaluated the changes of visual field and RNFL during 24 months follow-up in primary open-angle glaucoma (POAG) patients. Visual field and RNFL were detected by using GDx-VCC system and Octopus perimeter in 60 patients with POAG in early stage (60 eyes), 32 in moderate stage (32 eyes) and 30 in advanced stage (30 eyes). They found GDx-VCC system to be useful to evaluate the progression of POAG in early and moderate stage by detecting the change of the RNFL thickness with long-term follow-up. 
In a study by Medeiros et al. 213 eyes of 213 patients with an average follow-up time of 4.5 years were studied annually with the ECC and VCC, along with optic disc stereophotographs and SAP. Thirty-three patients (15%) showed progression over time on visual fields and/or stereophotographs. Mean rates of average RNFL thickness change measured by the GDx-ECC were significantly different in progressors versus nonprogressors (−1.24 μ/year vs. −0.34 μ/year; P < 0.001). AUROC curve for discriminating progressors from nonprogressors was significantly higher for rates of change measured by ECC compared to VCC (0.89 vs. 0.65; P <.001). Rates of RNFL change detected by the GDx ECC were significantly greater in eyes with progressive glaucoma compared to eyes with stable disease. Also, the ECC performed significantly better than the VCC for detection of change.
Makabe et al.  also found that longitudinal progression in NFI obtained with GDxVCC was significantly correlated with that in Humphrey field analyzer humphrey field analyzer (HFA) parameters, such as mean deviation and pattern standard deviation and recommended that GDx-VCC is a useful tool for longitudinal follow-up assessment of glaucoma.
| Special Scenarios|| |
Effect of cataract surgery
Scanning laser polarimetry measures the shift in polarization of light that is associated with the natural birefringence of RNFL. Other birefringent structures within the eye include cornea and lens. The GDx compensates for the birefringence of the cornea in part but that caused by lens is not compensated and hence there can be change in RNFL thickness after cataract surgery.
Vetrugno et al.  examined 68 eyes of 68 patients undergoing phacoemulsification with foldable intraocular lens (IOL) implantation (silicone and acrylic) by SLP before and after surgery. They could not find any statistically significant difference between SLP parameters before and after cataract surgery, regardless of the type of IOL implanted.
In contrast to this, Park et al.,  Iacono et al.  and Dada et al.  stressed on new baseline SLP measurements after cataract extraction with IOL. Gazzard et al.  have also shown a significant change in SLP measurements after cataract extraction, especially in patients with posterior subcapsular cataract.
Brittain et al.  investigated changes in GDx-VCC parameters caused by posterior capsular opacification. Typical scan score significantly improved after laser from 33 to 55.1 (P = 0.001; n = 19) and TSNIT score was significantly lower, dropping from 62.3 to 58.9 (P = 0.03; n = 19).
Effect of refractive surgery (laser in situ keratomileusis (LASIK), photo refractive keratectomy)
Dada et al.  and Zangwill et al.  could not find any statistically significant change in any of the parameters of RNFL before and after LASIK. Katsanos et al.,  however, found transient changes in RNFL by SLP after LASIK, which became stable by the 3 rd postoperative month.
| Comparison with other Diagnostic Modalities|| |
Scanning laser polarimetry versus optical coherence tomography
Leung et al.,  Chung  and Kim et al.  have reported that the total average RNFL thickness measured with Stratus OCT and GDx-VCC were highly correlated. Yoo et al. found that overall performance of Stratus OCT and GDx-VCC about their internal normative database was not significantly different.
Aptel et al.  tested 120 eyes of 120 patients (40 with healthy eyes, 40 with suspected glaucoma, and 40 with glaucoma) on Cirrus-OCT, GDx-VCC, and SAP. With spectral-domain OCT, the correlations (r(2)) between RNFL thickness and visual field sensitivity ranged from 0.082 to 0.726. By comparison, with GDx-VCC, the correlations ranged from 0.062 to 0.362. The largest AUROC were seen for OCT superior thickness (0.963 ± 0.022; P < 0.001) in eyes with glaucoma and for OCT average thickness (0.888 ± 0.072; P < 0.001) in eyes with suspected glaucoma. They concluded that the structure-function relationship was significantly stronger with OCT than with GDx-VCC.
Lee et al. found AUROCs of the Cirrus OCT to be significantly higher than those of GDx-VCC in glaucoma detection. The best-performing parameter was the NFI in GDx-VCC and inferior RNFL thickness in cirrus-OCT (AUROC = 0.912, 0.961, P = 0.045). There was good agreement between the two instruments with respect to abnormal classifications (kappa, 0.611-0.766).
Xu et al.  investigated the performance of SDOCT and SLP to detect progressive RNFL changes in glaucoma and found that at a comparable level of specificity, progressive RNFL thinning was detected more often than progressive reduction of retardance.
Bagga et al.  showed that images with ARPs have a weaker correlation with RNFL thickness measured with OCT compared with images with normal patterns. These results were subsequently confirmed in a study by Sehi et al. 
Garas  found that the diagnostic accuracy of the GDx-VCC/ECC NFI and RTVue-OCT average RNFLT were similar in glaucoma cases.
Schallenberg et al.  used GDx-VCC, GDx-ECC, and Spectralis-OCT to study RNFL measurements in 92 eyes of 92 glaucoma patients with various amount of glaucomatous damage and found that they cannot be directly compared because of differences in method of the devices.
| Scanning Laser Polarimetry Versus Frequency Doubling Technology/Standard Automated Perimetry|| |
In the Groningen Longitudinal Glaucoma Study  seventy glaucoma suspect patients with normal test results at baseline were followed prospectively for 4 years with SAP, frequency doubling technology and GDx. Conversion of suspects was better picked up by GDx.
Scanning laser polarimetry versus Heidelberg retina tomograph
Alencar et al.  found that the measurements of rates of change in GDx-VCC RNFL thickness were superior to HRT Rim area in identifying eyes with progression.
Kanamori et al.  also found GDx-VCC, HRT and OCT to be useful in preperimetric and early glaucoma. In their study of glaucomatous eyes with or without early visual field defects, SLP and OCT performed similarly or had better discriminating abilities compared with CSLO.
Medeiros et al.  compared the GDx-VCC, the HRT II and the Stratus OCT best parameters (NFI, the linear discriminant function and Inferior Average, respectively) in their ability to discriminate healthy eyes from glaucomatous patients. The AUROC curve of the three parameters indicated excellent discriminating power, without significant differences between them. For the GDx-VCC the three best performing parameters were NFI (0.91), Normalized Inferior Area (0.86) and TSNIT average (0.85).
| Scanning Laser Polarimetry Versus Matrix Perimetry, Optical Coherence Tomography, and Retinal Nerve Fiber Layer Photography using Heidelberg Retina Angiograph 1|| |
Hong et al.  compared the efficacy of detecting early glaucoma using all these in 72 POAG patients with early-stage visual field defects and 48 healthy controls. The AUROCs of Matrix perimetry, GDx-VCC, OCT, and RNFL photography using Heidelberg retina angiograph 1 with the best discriminating parameter were 0.990, 0.906, 0.794, and 0.751, respectively.
| Strengths of Scanning Laser Polarimetry|| |
Scanning laser polarimetry is a sophisticated technology that provides highly reproducible, objective measurements of the peripapillary RNFL. It is easy to operate, does not require pupillary dilatation, is highly reproducible, does not employ a reference plane, does not require correction for ocular magnification and incorporates an age-matched normative database. GDx-ECC has largely reduced ARP and is capable of neutralizing any residual anterior segment birefringence not completely compensated by VCC.
Retinal nerve fiber layer Integrity™
The RNFL-I measurement is the phase shift that occurs when polarized light passes through the RNFL. SLP captures birefringence, a tissue property that depends on the integrity of retinal ganglion cell axon microtubules and neurofilaments and hence SLP has the capacity not only to corroborate findings of RNFL thinning as determined by other methods, but may also provide insight into structural damage due to changes in density and orientation of the RNFL microstructures that may precede or occur in the absence of RNFL thinning.
| Limitations|| |
Image quality may be compromised if PPA falls under the measurement circle. Kunimatsu et al.  found reproducibility of GDx parameters to be similar among three circles in normal eyes (P > 0.05), whereas coefficients of reproducibility of TSNIT average (P = 0.006) and superior average (P = 0.035) were smaller in the smaller circles in OAG eyes. They recommended using medium circles in OAG patients as the obstructing influences of PPA on GDx measurement could be avoided.
Manufacturer recommendation is to avoid scans in eyes with refraction outside the + 5/−10 spherical diopter range as the probability of obtaining low-quality scans increases markedly. Patients wearing contact lenses can undergo GDx-VCC. Wang et al.  found that the average RNFL thickness measured with Cirrus HD-OCT decreases as the degree of myopia increases while no such correlation was detected in GDx-ECC. Wang et al.  in another study found RNFL thickness to be lower in all but the nasal quadrant in patients with POAG and high myopia (HM), compared to patients with only HM. GDx-VCC was found to be better than OCT in detecting POAG in HM patients. Dada et al.  studied RNFL thickness in myopia with SLP. Moderate myopes showed a significant thinning of the RNFL. In HM due to PPA and contribution of scleral birefringence, the RNFL values were abnormally high.
Yu et al.  and Bozkurt et al.  have reported GDx to be unreliable in tilted disc.
A large area centered on the fovea (6° × 6° square region) should be used in macular pathology.  Dada et al.  studied the effect of two macular birefringence protocols (bow-tie retardation and irregular macular scan) using GDx-VCC on RNFL thickness parameters and found the standard protocol to overestimate RNFL thickness in eyes with macular lesion which was normalized using the irregular macular pattern protocol.
| Conclusion|| |
GDx-VCC has emerged as a valuable, objective tool of clinical value in the management of glaucoma. GDx-ECC has overcome the presence of ARPs and improved the diagnostic accuracy and the relationship between functional and structural measurements. It is useful for early detection of structural damage to RNFL and longitudinal follow-up to pick up progression. However the clinician should be aware of its limitations, reconfirm an abnormal test, check the image quality and always use the GDx test report in conjunction with clinical evaluation of the optic nerve head and functional testing for starting or adjusting glaucoma therapy.
| References|| |
Sommer A, Katz J, Quigley HA, Miller NR, Robin AL, Richter RC, et al.
Clinically detectable nerve fiber atrophy precedes the onset of glaucomatous field loss. Arch Ophthalmol 1991;109:77-83.
Sommer A, Miller NR, Pollack I, Maumenee AE, George T. The nerve fiber layer in the diagnosis of glaucoma. Arch Ophthalmol 1977;95:2149-56.
Caprioli J, Prum B, Zeyen T. Comparison of methods to evaluate the optic nerve head and nerve fiber layer for glaucomatous change. Am J Ophthalmol 1996;121:659-67.
Quigley HA, Addicks EM, Green WR. Optic nerve damage in human glaucoma. III. Quantitative correlation of nerve fiber loss and visual field defect in glaucoma, ischemic neuropathy, papilledema, and toxic neuropathy. Arch Ophthalmol 1982;100:135-46.
Quigley HA, Katz J, Derick RJ, Gilbert D, Sommer A. An evaluation of optic disc and nerve fiber layer examinations in monitoring progression of early glaucoma damage. Ophthalmology 1992;99:19-28.
Airaksinen PJ, Alanko HI. Effect of retinal nerve fibre loss on the optic nerve head configuration in early glaucoma. Graefes Arch Clin Exp Ophthalmol 1983;220:193-6.
Quigley HA. Examination of the retinal nerve fiber layer in the recognition of early glaucoma damage. Trans Am Ophthalmol Soc 1986;84:920-66.
Quigley HA, Enger C, Katz J, Sommer A, Scott R, Gilbert D. Risk factors for the development of glaucomatous visual field loss in ocular hypertension. Arch Ophthalmol 1994;112:644-9.
Jonas JB, Hayreh SS. Localised retinal nerve fibre layer defects in chronic experimental high pressure glaucoma in rhesus monkeys. Br J Ophthalmol 1999;83:1291-5.
Zhou Q, Weinreb RN. Individualized compensation of anterior segment birefringence during scanning laser polarimetry. Invest Ophthalmol Vis Sci 2002;43:2221-8.
Weinreb RN, Dreher AW, Coleman A, Quigley H, Shaw B, Reiter K. Histopathologic validation of Fourier-ellipsometry measurements of retinal nerve fiber layer thickness. Arch Ophthalmol 1990;108:557-60.
Morgan JE, Waldock A, Jeffery G, Cowey A. Retinal nerve fibre layer polarimetry: Histological and clinical comparison. Br J Ophthalmol 1998;82:684-90.
Dreher AW, Reiter K. Retinal laser ellipsometry: A new method for measuring the retinal nerve fiber thickness distribution. Clin Vision Sci 1992;7:481-8.
Weinreb RN, Bowd C, Zangwill LM. Glaucoma detection using scanning laser polarimetry with variable corneal polarization compensation. Arch Ophthalmol 2003;121:218-24.
Medeiros FA, Zangwill LM, Bowd C, Bernd AS, Weinreb RN. Fourier analysis of scanning laser polarimetry measurements with variable corneal compensation in glaucoma. Invest Ophthalmol Vis Sci 2003;44:2606-12.
Shirakashi M, Yaoeda K, Fukushima A, Funaki S, Funaki H, Ofuchi N, et al
. Usefulness of GDx VCC in glaucoma detection. J Eye 2003;20:1019-21.
Greenfield DS, Knighton RW, Huang XR. Effect of corneal polarization axis on assessment of retinal nerve fiber layer thickness by scanning laser polarimetry. Am J Ophthalmol 2000;129:715-22.
Bowd C, Zangwill LM, Weinreb RN. Association between scanning laser polarimetry measurements using variable corneal polarization compensation and visual field sensitivity in glaucomatous eyes. Arch Ophthalmol 2003;121:961-6.
Saito H, Tomidokoro A, Yanagisawa M, Aihara M, Tomita G, Araie M. Scanning laser polarimetry with enhanced corneal compensation in patients with open-angle glaucoma. J Glaucoma 2008;17:24-9.
Medeiros FA, Bowd C, Zangwill LM, Patel C, Weinreb RN. Detection of glaucoma using scanning laser polarimetry with enhanced corneal compensation. Invest Ophthalmol Vis Sci 2007;48:3146-53.
Dada T, Gadia R, Aggarwal A, Dave V, Gupta V, Sihota R. Retinal nerve fiber layer thickness measurement by scanning laser polarimetry (GDxVCC) at conventional and modified diameter scans in normals, glaucoma suspects, and early glaucoma patients. J Glaucoma 2009;18:448-52.
Bagga H, Greenfield DS, Feuer W, Knighton RW. Scanning laser polarimetry with variable corneal compensation and optical coherence tomography in normal and glaucomatous eyes. Am J Ophthalmol 2003;135:521-9.
Schlottmann PG, De Cilla S, Greenfield DS, Caprioli J, Garway-Heath DF. Relationship between visual field sensitivity and retinal nerve fiber layer thickness as measured by scanning laser polarimetry. Invest Ophthalmol Vis Sci 2004;45:1823-9.
Reus NJ, Lemij HG. Scanning laser polarimetry of the retinal nerve fiber layer in perimetrically unaffected eyes of glaucoma patients. Ophthalmology 2004;111:2199-203.
Choplin NT, Zhou Q, Knighton RW. Effect of individualized compensation for anterior segment birefringence on retinal nerve fiber layer assessments as determined by scanning laser polarimetry. Ophthalmology 2003;110:719-25.
Langlotz CP. Fundamental measures of diagnostic examination performance: Usefulness for clinical decision making and research. Radiology 2003;228:3-9.
Sanchez-Galeana C, Bowd C, Blumenthal EZ, Gokhale PA, Zangwill LM, Weinreb RN. Using optical imaging summary data to detect glaucoma. Ophthalmology 2001;108:1812-8.
Rao HL, Venkatesh CR, Vidyasagar K, Yadav RK, Addepalli UK, Jude A, et al.
Retinal nerve fiber Layer measurements by scanning laser Polarimetry with enhanced corneal compensation in healthy subjects. J Glaucoma 2013.
Medeiros FA, Vizzeri G, Zangwill LM, Alencar LM, Sample PA, Weinreb RN. Comparison of retinal nerve fiber layer and optic disc imaging for diagnosing glaucoma in patients suspected of having the disease. Ophthalmology 2008;115:1340-6.
Mohammadi K, Bowd C, Weinreb RN, Medeiros FA, Sample PA, Zangwill LM. Retinal nerve fiber layer thickness measurements with scanning laser polarimetry predict glaucomatous visual field loss. Am J Ophthalmol 2004;138:592-601.
Badalà F, Nouri-Mahdavi K, Raoof DA, Leeprechanon N, Law SK, Caprioli J. Optic disk and nerve fiber layer imaging to detect glaucoma. Am J Ophthalmol 2007;144:724-32.
Shaikh A, Salmon JF. The role of scanning laser polarimetry using the GDx variable corneal compensator in the management of glaucoma suspects. Br J Ophthalmol 2006;90:1454-7.
Li G, Fansi AK, Harasymowycz P. Screening for glaucoma using GDx-VCC in a population with=1 risk factors. Can J Ophthalmol 2013;48:279-85.
Benítez-del-Castillo J, Martinez A, Regi T. Correlation between scanning laser polarimetry with and without enhanced corneal compensation and high-definition optical coherence tomography in normal and glaucomatous eyes. Int J Clin Pract 2011;65:807-16.
Zheng W, Baohua C, Qun C, Zhi Q, Hong D. Retinal nerve fiber layer images captured by GDx-VCC in early diagnosis of glaucoma. Ophthalmologica 2008;222:17-20.
Da Pozzo S, Fuser M, Vattovani O, Di Stefano G, Ravalico G. GDx-VCC performance in discriminating normal from glaucomatous eyes with early visual field loss. Graefes Arch Clin Exp Ophthalmol 2006;244:689-95.
Parikh RS, Parikh SR, Kumar RS, Prabakaran S, Babu JG, Thomas R. Diagnostic capability of scanning laser polarimetry with variable cornea compensator in Indian patients with early primary open-angle glaucoma. Ophthalmology 2008;115:1167-1172.e1.
Fortune B, Burgoyne CF, Cull G, Reynaud J, Wang L. Onset and progression of peripapillary retinal nerve fiber layer (RNFL) retardance changes occur earlier than RNFL thickness changes in experimental glaucoma. Invest Ophthalmol Vis Sci 2013;54:5653-61.
Reus NJ, Lemij HG. Diagnostic accuracy of the GDx VCC for glaucoma. Ophthalmology 2004;111:1860-5.
Henderson PA, Medeiros FA, Zangwill LM, Weinreb RN. Relationship between central corneal thickness and retinal nerve fiber layer thickness in ocular hypertensive patients. Ophthalmology 2005;112:251-6.
Choi J, Cho HS, Lee CH, Kook MS. Scanning laser polarimetry with variable corneal compensation in the area of apparently normal hemifield in eyes with normal-tension glaucoma. Ophthalmology 2006;113:1954-60.
Jung JI, Kim JH, Kook MS. Comparison of retinal nerve fiber layer measurements between NTG and HTG using GDx-VCC. Korean J Ophthalmol 2006;20:26-32.
Zareii R, Soleimani M, Moghimi S, Eslami Y, Fakhraie G, Amini H. Relationship between GDx VCC and Stratus OCT in juvenile glaucoma. Eye (Lond) 2009;23:2182-6.
Dimopoulos AT, Katsanos A, Mikropoulos DG, Giannopoulos T, Empeslidis T, Teus MA, et al.
Scanning laser polarimetry in eyes with exfoliation syndrome. Eur J Ophthalmol 2013;23:743-50.
Tóth M, Holló G. Enhanced corneal compensation for scanning laser polarimetry on eyes with atypical polarisation pattern. Br J Ophthalmol 2005;89:1139-42.
Da Pozzo S, Marchesan R, Canziani T, Vattovani O, Ravalico G. Atypical pattern of retardation on GDx-VCC and its effect on retinal nerve fibre layer evaluation in glaucomatous eyes. Eye (Lond) 2006;20:769-75.
Bagga H, Greenfield DS, Feuer WJ. Quantitative assessment of atypical birefringence images using scanning laser polarimetry with variable corneal compensation. Am J Ophthalmol 2005;139:437-46.
Mai TA, Reus NJ, Lemij HG. Diagnostic accuracy of scanning laser polarimetry with enhanced versus variable corneal compensation. Ophthalmology 2007;114:1988-93.
Orlev A, Horani A, Rapson Y, Cohen MJ, Blumenthal EZ. Clinical characteristics of eyes demonstrating atypical patterns in scanning laser polarimetry. Eye (Lond) 2008;22:1378-83.
Reus NJ, Zhou Q, Lemij HG. Enhanced imaging algorithm for scanning laser polarimetry with variable corneal compensation. Invest Ophthalmol Vis Sci 2006;47:3870-7.
Tóth M, Holló G. Evaluation of enhanced corneal compensation in scanning laser polarimetry: Comparison with variable corneal compensation on human eyes undergoing LASIK. J Glaucoma 2006;15:53-9.
Bowd C, Tavares IM, Medeiros FA, Zangwill LM, Sample PA, Weinreb RN. Retinal nerve fiber layer thickness and visual sensitivity using scanning laser polarimetry with variable and enhanced corneal compensation. Ophthalmology 2007;114:1259-65.
Hoesl LM, Tornow RP, Schrems WA, Horn FK, Mardin CY, Kruse FE, et al.
Glaucoma diagnostic performance of GDxVCC and spectralis OCT on eyes with atypical retardation pattern. J Glaucoma 2013;22:317-24.
Wang Z, Liu XW, Li XY, Zhang WJ, Dai H. Detection of the changes of retinal nerve fiber layer thickness by GDx-VCC laser scanning polarimetry in primary open angle glaucoma patients. Zhonghua Yan Ke Za Zhi 2012;48:497-501.
Medeiros FA, Zangwill LM, Alencar LM, Sample PA, Weinreb RN. Rates of progressive retinal nerve fiber layer loss in glaucoma measured by scanning laser polarimetry. Am J Ophthalmol 2010;149:908-15.
Makabe K, Takei K, Oshika T. Longitudinal relationship between retinal nerve fiber layer thickness parameters assessed by scanning laser polarimetry (GDxVCC) and visual field in glaucoma. Graefes Arch Clin Exp Ophthalmol 2012;250:575-81.
Vetrugno M, Trabucco T, Sisto D, Sborgia C. Effect of cataract surgery and foldable intraocular lens implantation on retinal nerve fiber layer as measured by scanning laser polarimetry with variable corneal compensator. Eur J Ophthalmol 2004;14:106-10.
Park RJ, Chen PP, Karyampudi P, Mills RP, Harrison DA, Kim J. Effects of cataract extraction with intraocular lens placement on scanning laser polarimetry of the peripapillary nerve fiber layer. Am J Ophthalmol 2001;132:507-11.
Iacono P, Da Pozzo S, Vattovani O, Tognetto D, Ravalico G. Scanning laser polarimetry of nerve fiber layer thickness in normal eyes after cataract phacoemulsification and foldable intraocular lens implantation. J Cataract Refract Surg 2005;31:1042-9.
Dada T, Behera G, Agarwal A, Kumar S, Sihota R, Panda A. Effect of cataract surgery on retinal nerve fiber layer thickness parameters using scanning laser polarimetry (GDxVCC). Indian J Ophthalmol 2010;58:389-94.
Gazzard G, Foster PJ, Devereux JG, Oen F, Chew PT, Khaw PT, et al.
Effect of cataract extraction and intraocular lens implantation on nerve fibre layer thickness measurements by scanning laser polarimeter (GDx) in glaucoma patients. Eye (Lond) 2004;18:163-8.
Brittain CJ, Fong KC, Hull CC, Gillespie IH. Changes in scanning laser polarimetry before and after laser capsulotomy for posterior capsular opacification. J Glaucoma 2007;16:112-6.
Dada T, Chaudhary S, Muralidhar R, Nair S, Sihota R, Vajpayee RB. Evaluation of retinal nerve fiber layer thickness measurement following laser in situ
keratomileusis using scanning laser polarimetry. Indian J Ophthalmol 2007;55:191-4.
Zangwill LM, Abunto T, Bowd C, Angeles R, Schanzlin DJ, Weinreb RN. Scanning laser polarimetry retinal nerve fiber layer thickness measurements after LASIK. Ophthalmology 2005;112:200-7.
Katsanos A, Kóthy P, Nagy ZZ, Holló G. Scanning laser polarimetry of retinal nerve fibre layer thickness after laser assisted in situ
keratomileusis (LASIK): Stability of the values after the third post-LASIK month. Acta Physiol Hung 2004;91:119-30.
Leung CK, Chan WM, Chong KK, Yung WH, Tang KT, Woo J, et al.
Comparative study of retinal nerve fiber layer measurement by StratusOCT and GDx VCC, I: Correlation analysis in glaucoma. Invest Ophthalmol Vis Sci 2005;46:3214-20.
Chung YS, Sohn YH. The relationship between optical coherence tomography and scanning laser polarimetry measurements in glaucoma. Korean J Ophthalmol 2006;20:225-9.
Kim HG, Heo H, Park SW. Comparison of scanning laser polarimetry and optical coherence tomography in preperimetric glaucoma. Optom Vis Sci 2011;88:124-9.
Yoo YC, Park KH. Comparison of optical coherence tomography and scanning laser polarimetry for detection of localized retinal nerve fiber layer defects. J Glaucoma 2010;19:229-36.
Aptel F, Sayous R, Fortoul V, Beccat S, Denis P. Structure-function relationships using spectral-domain optical coherence tomography: Comparison with scanning laser polarimetry. Am J Ophthalmol 2010;150:825-33.
Lee S, Sung KR, Cho JW, Cheon MH, Kang SY, Kook MS. Spectral-domain optical coherence tomography and scanning laser polarimetry in glaucoma diagnosis. Jpn J Ophthalmol 2010;54:544-9.
Xu G, Weinreb RN, Leung CK. Retinal nerve fiber layer progression in glaucoma: A comparison between retinal nerve fiber layer thickness and retardance. Ophthalmology 2013;120:2493-500.
Sehi M, Ume S, Greenfield DS, Scanning laser polarimetry with enhanced corneal compensation and optical coherence tomography in normal and glaucomatous eyes. Invest Ophthalmol Vis Sci 2007;48:2099-104.
Garas A, Vargha P, Holló G. Comparison of diagnostic accuracy of the RTVue Fourier-domain OCT and the GDx-VCC/ECC polarimeter to detect glaucoma. Eur J Ophthalmol 2012;22:45-54.
Schallenberg M, Dekowski D, Kremmer S, Selbach JM, Steuhl KP. Comparison of Spectralis-OCT, GDxVCC and GDxECC in assessing retinal nerve fiber layer (RNFL) in glaucomatous patients. Graefes Arch Clin Exp Ophthalmol 2013;251:1343-53.
Jansonius NM, Heeg GP. The Groningen Longitudinal Glaucoma Study. II. A prospective comparison of frequency doubling perimetry, the GDx nerve fibre analyser and standard automated perimetry in glaucoma suspect patients. Acta Ophthalmol 2009;87:429-32.
Alencar LM, Zangwill LM, Weinreb RN, Bowd C, Sample PA, Girkin CA, et al.
A comparison of rates of change in neuroretinal rim area and retinal nerve fiber layer thickness in progressive glaucoma. Invest Ophthalmol Vis Sci 2010;51:3531-9.
Kanamori A, Nagai-Kusuhara A, Escaño MF, Maeda H, Nakamura M, Negi A. Comparison of confocal scanning laser ophthalmoscopy, scanning laser polarimetry and optical coherence tomography to discriminate ocular hypertension and glaucoma at an early stage. Graefes Arch Clin Exp Ophthalmol 2006;244:58-68.
Medeiros FA, Zangwill LM, Bowd C, Weinreb RN. Comparison of the GDx VCC scanning laser polarimeter, HRT II confocal scanning laser ophthalmoscope, and stratus OCT optical coherence tomograph for the detection of glaucoma. Arch Ophthalmol 2004;122:827-37.
Hong S, Ahn H, Ha SJ, Yeom HY, Seong GJ, Hong YJ. Early glaucoma detection using the Humphrey Matrix Perimeter, GDx VCC, Stratus OCT, and retinal nerve fiber layer photography. Ophthalmology 2007;114:210-5.
Kunimatsu S, Tomidokoro A, Saito H, Aihara M, Tomita G, Araie M. Performance of GDx VCC in eyes with peripapillary atrophy: Comparison of three circle sizes. Eye (Lond) 2008;22:173-8.
Wang G, Qiu KL, Lu XH, Sun LX, Liao XJ, Chen HL, et al.
The effect of myopia on retinal nerve fibre layer measurement: A comparative study of spectral-domain optical coherence tomography and scanning laser polarimetry. Br J Ophthalmol 2011;95:255-60.
Wang Z, Liu XW, Li XY, Zhang WJ, Dai H. Detection of the changes of retinal nerve fiber layer thickness by GDx-VCC laser scanning polarimetry in primary open angle glaucoma patients. Zhonghua Yan Ke Za Zhi 2012;48:497-501.
Dada T, Aggarwal A, Bali SJ, Sharma A, Shah BM, Angmo D, et al.
Evaluation of retinal nerve fiber layer thickness parameters in myopic population using scanning laser polarimetry (GDxVCC). Nepal J Ophthalmol 2013;5:3-8.
Yu S, Tanabe T, Hangai M, Morishita S, Kurimoto Y, Yoshimura N. Scanning laser polarimetry with variable corneal compensation and optical coherence tomography in tilted disk. Am J Ophthalmol 2006;142:475-82.
Bozkurt B, Irkeç M, Tatlipinar S, Erdener U, Orhan M, Gedik S, et al.
Retinal nerve fiber layer analysis and interpretation of GDx parameters in patients with tilted disc syndrome. Int Ophthalmol 2001;24:27-31.
Bagga H, Greenfield DS, Knighton RW. Scanning laser polarimetry with variable corneal compensation: Identification and correction for corneal birefringence in eyes with macular disease. Invest Ophthalmol Vis Sci 2003;44:1969-76.
Dada T, Tinwala SI, Dave V, Agarwal A, Sharma R, Wadhwani M. Effect of change in macular birefringence imaging protocol on retinal nerve fiber layer thickness parameters using GDx VCC in eyes with macular lesions. Int Ophthalmol 2014;34:901-7.
[Figure 1], [Figure 2], [Figure 3], [Figure 4], [Figure 5], [Figure 6]