|Year : 2019 | Volume
| Issue : 10 | Page : 1624-1627
Light exposure from microscope versus intracameral illumination during cataract surgery
Yu Jeong Kim1, Dong Heun Nam2, Young Jae Kim3, Kwang Gi Kim3, Seong-Woo Kim4, Tae-Young Chung5, Sung Jin Lee6, Kyu Hyung Park7
1 Department of Ophthalmology, Gangneung Asan Medical Center, Gangneung, Korea
2 Department of Ophthalmology, Gachon University Gil Hospital, Incheon, Korea
3 Department of Biomedical Engineering, Gachon University College of Medicine, Incheon, Korea
4 Department of Ophthalmology, Korea University College of Medicine, Seoul, Korea
5 Department of Ophthalmology, Samsung Medical Center, Sungkyunkwan University School of Medicine, Korea
6 Department of Ophthalmology, Soonchunhyang University College of Medicine, Soonchunhyang University Hospital, Korea
7 Department of Ophthalmology, Seoul National University College of Medicine, Seoul National University Bundang Hospital, Seongnam, Korea
|Date of Submission||14-Feb-2019|
|Date of Acceptance||01-Jun-2019|
|Date of Web Publication||23-Sep-2019|
Prof. Dong Heun Nam
Department of Ophthalmology, Gachon University Gil Hospital, 21, Namdong-daero 774 Beon-gil, Namdong-gu, Incheon - 21565
Source of Support: None, Conflict of Interest: None
Purpose: To evaluate light exposure from microscope versus intracameral illuminations to patient's and surgeon's retina during cataract surgery. Methods: Thirty consecutive patients who had cataract surgery using microscope and intracameral illuminations. At the point of the ocular of an operating microscope, optical illuminance and irradiance from the microscope illumination (60, 40, 20% intensity) and the intracameral illumination (60% intensity) were measured using a light meter and a spectrometer at a pause after lens capsule polishing in cataract surgery. Results: Average illuminance (lux) was 1.46, 0.66, 0.27, and 0.1 from 60%, 40%, 20% intensity microscope illuminations and 60% intracameral illumination. Average total spectral irradiance (μW/cm2) was 1.25, 0.65, 0.26, and 0.03 from 60%, 40%, 20% intensity microscope illuminations and 60% intracameral illumination. Conclusion: Microscope ocular illuminance and irradiance during cataract surgery were higher in the microscope illumination than in the intracameral illumination. It suggests that light exposure reaching patient's and surgeon's retina during cataract surgery is lower in the intracameral illumination than in the microscope illumination.
Keywords: Cataract surgery, illuminance, intracameral illumination, irradiance, light exposure, phototoxicity
|How to cite this article:|
Kim YJ, Nam DH, Kim YJ, Kim KG, Kim SW, Chung TY, Lee SJ, Park KH. Light exposure from microscope versus intracameral illumination during cataract surgery. Indian J Ophthalmol 2019;67:1624-7
|How to cite this URL:|
Kim YJ, Nam DH, Kim YJ, Kim KG, Kim SW, Chung TY, Lee SJ, Park KH. Light exposure from microscope versus intracameral illumination during cataract surgery. Indian J Ophthalmol [serial online] 2019 [cited 2019 Dec 13];67:1624-7. Available from: http://www.ijo.in/text.asp?2019/67/10/1624/267438
A red reflex, produced by reflection of coaxial light from the macula back to the observer, is one of the most important features of an ophthalmic microscope for cataract surgery. However, it has long been known that the red reflex in cataract surgery can induce retinal phototoxicity.
It is important to find ways to reduce retinal phototoxicity during cataract surgery. For preventing the toxicity, cataract surgeons try to minimize the level of microscope illumination, tilt the microscope beam away from the fovea or shorten the surgical time. In view of maximizing operational safety, which calls for sufficient red reflex during cataract surgery, nonetheless, these modifications cannot be considered routine.,
An advanced cataract surgery technique using the intracameral illumination has been introduced with an enhanced 3D effect and an improved depth perception of lens.,,,, Increasing clinical experience with the intracameral illumination has provided us with a better understanding of certain features of this new technology. With concern about the retinal phototoxicity from the microscope coaxial light, this technique using the oblique light focused onto the lens offers the advantage of reduced direct illumination of the macula. There have been no studies to date that show the measurement of illuminance and irradiance reaching patient's or surgeon's retina during cataract surgery. Therefore, we performed a prospective comparative indirect observation to assess light exposure to patient's and surgeon's retina between microscope versus intracameral illuminations during phacoemulsification cataract surgery [Figure 1].
|Figure 1: (a) Microscope illumination versus (b) Intracameral illumination. A red reflex is produced by reflection of bright coaxial light from the macula back to the observer (the ocular of an operating microscope) (a). Illumination, while, a side light of intracameral illumination is shining into not macula but peripheral retina. Therefore, the light reflection from the macula back to the observer is decreased (b)|
Click here to view
| Methods|| |
Gachon University IRB (GAIRB2016-65) approved this study before its initiation. Subjects participated with full, informed consent and the study adhered to the tenets of the Declaration of Helsinki. This investigation took place from April 2017 to June 2017 at the Department of Ophthalmology, Gachon University Gil Hospital, Incheon, South Korea. A review of medical records was performed in which demographic data, medical history, ocular history, best-corrected visual acuity (BCVA), slit-lamp examination including lens LOCS III (Lens Opacities Classification System, Version III) scoring, dilated fundus examination, and fundus photography were collected and reviewed.
Inclusion criteria were patient age greater than 40 years and surgical indication of senile cataracts. Exclusion criteria were patients with complicated cataracts (e.g., small pupil, mature lens, prior trauma, or zonular weakness), previous intraocular surgery, or surgical time longer than 25 min. All procedures were performed by one physician (D.H.N.) with an operating microscope coaxial illumination (M844 C40, Leica Microsystems, Wetzlar, Germany) (microscope illumination) or a 23-gauge endoillumination (Xenon BrightStar, DORC, Ziudland, Netherlands) (intracameral illumination). The surgical technique described in our previous studies 5--8 was used in all cataract surgeries including the advanced lens capsule polishing. To optimize visualization at all stages of the cataract surgery, we changed and customized illumination according to the varying requirements of different surgical procedures and eyes.
Illuminance and spectral irradiance of each wavelength were measured using a light meter (TES1332A, Electrical Electronic Corp., Taiwan) and a spectrometer (FLAME-S, Ocean Optics, Inc., USA). The measurements were performed at the point of the ocular (eyepiece) of an operating microscope [Figure 2]. The illuminance and spectral irradiance from the microscope illumination (60, 40, 20% intensity) and the intracameral illumination (60% intensity) were measured at a pause after lens capsule polishing in cataract surgery with operating room lights off [Figure 3]. Total spectral irradiance (μW/cm 2) was calculated as the average integration value for each waveform from 30 eyes. The total surgical time including the illuminance and irradiance measurement time was compared.
|Figure 2: At the point of the ocular (eyepiece) of an operating microscope, illuminance and spectral irradiance of each wavelength were measured using a light meter (a) and a spectrometer (b)|
Click here to view
|Figure 3: At a pause after lens capsule polishing with operating room lights off, we measured under microscope illumination 60% intensity, which is good illumination (a), 40%, which is fair illumination (b), 20%, which is poor illumination (c), and endoilluminator 60% intensity, which is good illumination (d)|
Click here to view
All statistical analyses were performed using SPSS software (version 12.0, SPSS, Inc.). ANOVA test were conducted for univariate analysis to demonstrate significance of differences between microscope illumination and intracameral illumination. P values of 0.05 or less were considered statistically significant.
| Results|| |
Thirty consecutive patients participated in the study. No patient experienced intraoperative complication such as posterior capsule rupture or zonular dialysis. The patient demographics are described in [Table 1].
The total surgical time was 20.37 ± 1.33 min in microscope illumination group and 20.43 ± 1.45 min in intracameral illumination group (P = 0.624). Average illuminance (lux) was 1.46, 0.66, 0.27, and 0.10 from 60%, 40%, 20% intensity microscope illuminations and 60% intracameral illumination. The illuminance of 60% intracameral illumination was decreased to 6.8%, 15.1%, and 37.0% of levels of 60%, 40%, and 20% intensity microscope illuminations, respectively (P < 0.001, ANOVA test) [Figure 4].
|Figure 4: Average illuminance (lux) was 1.46, 0.66, 0.27, and 0.10 from 60%, 40%, 20% intensity microscope illuminations and 60% intracameral illumination. The illuminance was significantly different between the illuminations (P < 0.001, ANOVA test)|
Click here to view
Average total spectral irradiance (μW/cm 2) was 1.25, 0.65, 0.26, and 0.03 from 60%, 40%, 20% intensity microscope illuminations and 60% intracameral illumination. The total spectral irradiance of 60% intracameral illumination was decreased to 2.4%, 4.6%, and 11.5% of levels of 60%, 40%, and 20% intensity microscope illuminations, respectively (P < 0.001, ANOVA test) [Figure 5].
|Figure 5: Total spectral irradiance (μW/cm2) was calculated as the average integration value for each waveform from 30 eyes. B. Average total spectral irradiance (μW/cm2) was 1.25, 0.65, 0.26, and 0.03 from 60%, 40%, 20% intensity microscope illuminations and 60% intracameral illumination. The total spectral irradiance was significantly different between the illuminations (P < 0.001, ANOVA test)|
Click here to view
| Discussion|| |
Although intraoperative iatrogenic retinal phototoxicity is rare, it results in permanent visual dysfunction. In 1995, public health advisory from the US Food and Drug Administration was reported about Retinal Photic Injuries From Operating Microscopes During Cataract Surgery. The operating microscope manufacturers have since incorporated ultraviolet and infrared filters to minimize the possibility of light damage. However, intraoperative microscope light-induced photochemical damage can be caused by radiation between 400 and 550 nm, still opening the possibility of photochemical insult to the retina. Furthermore, the manufacturers do not provide a recommended illumination intensity level, leaving the surgeon free to set the intraoperative illumination intensity level according to individual preference. Therefore, it is very important for cataract surgeons to balance efficiency during cataract surgery, which calls for sufficient intraoperative illumination/visualization, and achieving good postoperative visual outcomes, which calls for the suppression of retinal phototoxicity. Raising the intensity of light to improve visibility leads to a sacrifice in safety. This study showed that the advanced cataract surgery technique using the intracameral illumination could be a viable option to achieve both operational efficiency and safety. The novel illumination not only improves observational conditions but also decreases the light exposure reaching retina during cataract surgery.
In terms of efficiency, the intracameral illumination enhances the details of the lens structures, mainly posterior capsule. In addition, it creates an enhanced 3D effect and improves depth perception of lens. The advantages could decrease the risk of intraoperative complications, especially in junior cataract surgeons.,,,,,
This is the first time intraoperative observation has been used to measure the light exposure during cataract surgery. Strengths of this study are its prospective nature and an intraindividual comparative design between microscope versus intracameral illuminations. This study suggests that light exposure reaching patient's and surgeon's retina during cataract surgery is much lower in the intracameral illumination than in the microscope illumination. It is consistent with previous study. However, this study did not predict the absolute intensity of light that the retina could become exposed to during cataract surgery in real life. This is because our study did not measure the intensity of the light on the macula but at the point of the ocular of an operating microscope. Therefore, it is difficult to compare the light values obtained from this observation to the toxic values reported in the literature.
| Conclusion|| |
In conclusion, this study highlights the utility of the intracameral illumination to minimize intraoperative light exposure reaching patient's and surgeon's retina. The advanced cataract surgery with reduced light exposure enabled excellent intraoperative visualization of lens structures. Considering reduction of potential phototoxicity, control of light intensity is an important factor. In view of patient or surgeon retinal phototoxicity, the intracameral illumination may be safer than the microscope illumination. To confirm the efficacy and safety of this emerging technology, however, it is necessary to perform a prospective study comparing the clinical outcomes of the novel cataract surgery with intracameral illumination versus those of the standard cataract surgery with microscope illumination.
Financial support and sponsorship
Conflicts of interest
There are no conflicts of interest.
| References|| |
Azzolini C, Brancato R, Venturi G, Bandello F, Pece A, Santoro P. Updating on intraoperative light-induced retinal injury. Int Ophthalmol 1995;18:269-76.
Wolffe M. How safe is the light during ophthalmic diagnosis and surgery. Eye 2016;30:186-8.
Harman FE, Corbett MC, Stevens JD. Effect of the angle of the operating microscope light beam on visual recovery after phacoemulsification: Randomized trial. J Cataract Refract Surg 2010;36:1311-5.
Kim BH. Lightless cataract surgery using a near-infrared operating microscope. J Cataract Refract Surg 2006;32:1683-90.
Moon H, Lee JH, Lee JY, Kim KH, Lee DY, Nam DH. Intracameral dynamic spotlight-assisted cataract surgery in eyes with corneal opacity, small pupil or advanced cataract. Acta Ophthalmol 2015;93:388-90.
Lee JY, Yoon J, Kim IN, Kim KH, Lee DY, Nam DH. Intracameral illuminator-guided advanced lens capsule polishing during cataract surgery in eyes with diabetic retinopathy. Retina 2012;32:1420-3.
Jung Y, Kim IN, Yoon J, Lee JY, Kim KH, Lee DY, et al
. Intracameral illuminator-assisted advanced cataract surgery combined with 23-gauge vitrectomy in eyes with poor red reflex. J Cataract Refract Surg 2013;39:845-50.
Wi J, Seo H, Lee JY, Nam DH. Phacoemulsification using a chisel-shaped illuminator: Enhanced depth trench, one-shot crack, and phaco cut. Eur J Ophthalmol 2016;26:279-80.
Yepez JB, Murati FA, Garcia F, Calderon V, de Yepez JC, Arevalo JF. Phacoemulsification outcomes with different illumination techniques. Eur J Ophthalmol 2017;27:797-800.
Seo H, Nam DH, Lee JY, Park SJ, Kim YJ, Kim SW, et al
. Macular photostress and visual experience between microscope and intracameral illumination during cataract surgery. J Cataract Refract Surg 2018;44:190-7.
FDA Public Health Advisory: Retinal Photic Injuries from operating microscopes during cataract surgery. FDA, 1995.
Michael R, Wegener A. Estimation of safe exposure time from an ophthalmic operating microscope with regard to ultraviolet radiation and blue-light hazards to the eye. J Opt Soc Am A Opt Image Sci Vis 2004;21:1388-92.
Park YM, Park JY, Lee JS, Lee JE, Byon IS, Park SW. Comparative analysis of brilliant blue G and an intracameral illuminator in assisting visualization of the anterior capsule in eyes with vitreous hemorrhage. J Cataract Refract Surg 2016;42:1015-21.
Inoue M, Uchida A, Shinoda K, Taira Y, Noda T, Ohnuma K, et al
. Images created in a model eye during simulated cataract surgery can be the basis for images perceived by patients during cataract surgery. Eye 2014;28:870-9.
[Figure 1], [Figure 2], [Figure 3], [Figure 4], [Figure 5]